1-Wire UNI sensors enables setting and calibration of the measured values, including an option to change the unit type of the value, an exponent and, based on the sensor's type, up to 8-point calibration chart. This allows us to connect various sensors with voltage or current output to Poseidon units via the 1Wire UNI bus.
Many industrial sensors have 0-10V, eventually 0(4)-20mA outputs. A typical example of this sensor is a clamp meter for measuring the AC current with magnetic induction (30A Current probe 1W-UNI sensor) or pressure meters.
If you then need to connect this sensor to a Poseidon unit (1-Wire UNI compatible), you can simply use Sensor 60V 1W-UNI v2 or Sensor 0-20mA 1W-UNI converters. From this point, Poseidon shows measured values in volts (V) or mA.
There are also conversion charts for getting the real value, however this is usually not a very practical solution.
HWg-Calibrator software therefore allows an implementation of any conversion chart directly to 1-Wire UNI converters.
Older converters Sensor 15V 1W-UNI, Sensor 60V 1W-UNI and Sensor 20mA 1W-UNI allows 3-point conversion (calibration), newer Sensor 60V 1W-UNI v2 and Sensor 0-20mA 1W-UNI can use an 8-point conversion chart.